Next: The Bench Acquisition TV and Fiber Back Illuminator
Previous: The Fiber Module
Up: THE BENCH SPECTROGRAPH
There are four filter slots on the Fiber Module. The first slot (A) holds aperture masks and is not currently used; the second slot (B) holds thick (0.35 in) filters (the interference filters for the Echelle grating); the other two (C and D) hold thinner (0.25 in) filters (the glass filters). Hence, it is possible to use a broad-band glass filter in concert with another glass filter or with an interference filter for maximal blocking.
A list of available filters for order separation is given in Table 3.
Table 3: List of Bench Spectrograph filters.
In that Table, LP stands for longpass, BP for bandpass. ``-" and ``+" are the half maxima. Transmission curves for the available filters are shown in Figures 3 and 4.
Figure 3: Transmission curves for the available glass filters. Note the red leak in the G6 filter. The glitches at 8600 Å are artifacts from the scanning process.
Figure 4: Transmission curves for the interference filters.